Each Fluke product is warranted to be free from defects in material and The A Manual Set provides complete information for operators and service or. Warning If the A Calibrator is operated in any way not specified by the A Operators Manual or other documentation provided by Fluke, protection. View and Download FLUKE A operator’s manual online. Multi-Product Calibrator. A Test Equipment pdf manual download.
|Published (Last):||6 September 2009|
|PDF File Size:||16.94 Mb|
|ePub File Size:||20.97 Mb|
|Price:||Free* [*Free Regsitration Required]|
Allow the A reading to stabilize.
Don’t have an account? Command Summary by Function Enter the corrected rms reading via the Calibrator Mainframe front panel keypad, then press enter Note The Calibrator Mainframe will warn when the entered value is out of bounds.
To access calibration steps described in paragraphs throughsimply send the command: Measure the baseline of each output fuke the corresponding topline measurement. The memory holds calibration constants and setup parameters.
The default setting is 25 mV 1 MHz. The Time Marker Function Inguard Supplies The inguard supplies are located on the Voltage assembly A8. Setting Voltage and Current Limits To set voltage and current entry limits, proceed as follows: Other suspects on the A6 assembly include relay K3 and resistor network Z5. Program the digital storage oscilloscope to the parameters listed below.
5500A Operator Manual
Then use the next sections to verify the DC Flke function. Adjusting the Rise Time for the Edge Function This procedure adjusts the edge rise time, and must be performed after repair. Wave Generator Verification at 50 O.
Removing the Keyboard and Accessing the Output Block To remove the keyboard and access the output block, proceed as follows: Overload Measurement Specifications Table Volt Specifications Table Changes in temperature sensors changes the output to 0 C.
Fluke Calculator A/COIL User Guide |
The 20 ms to 2 ns markers are generated on the A50 board. Remove the two front handles by removing the six Allen screws from the handles. Phase Specifications A Operator Manual Enter the LCR reading into the A when prompted.
Accessing the Fuse and Selecting Line Voltage Watts uncertainty power voltage You may find that you can place the second harmonic at dBc but the third harmonic is less than dBc. It uses discrete tluke and capacitors as references, with an amplifier in series.
Boost command Activate or deactivate a A Amplifier and set the source preference to the A Amplifier or to the A Calibrator. Page The minimum use specifications are usually determined by maintaining a specified test uncertainty ratio between the calibration equipment and the unit under test.
Page 63 Introduction Sets the output of the A and establishes a new reference point for the error mode. Table shows the test point. Manually lock the HP A to the range that gives the most resolution for the baseline measurements. Pulse Response Calibration Using a Tunnel Diode Pulser You can use the calibrator to drive a tunnel diode pulser Fluke Part Numberor Tektronixor equivalentallowing you to check for pulse edge rise times as fast as ps.
Measure the topline first. Use copper hookup wire for all connections, except for steps 17 to When one output is a harmonic of the other, the phase shift is based on the phase angle or power factor cosine of the harmonic signal.
If a specified instrument is not available, you can substitute an instrument that assures a 4: Then follow these steps to calibrate edge amplitude:.
You may need to adjust the signal averaging on the IB. Diagnostics Error Flhke If an error message appears during diagnostics, check the following annotated list to determine which assembly, and what circuit, is suspect.
The highest fluk voltage is limited to 2. Some commands require a previous condition be set before the command will be accepted by the A Calibrator.
Fluke does not warrant that software will be error free or operate without interruption. Figure is a block diagram of the current function.
FLUKE A OPERATOR’S MANUAL Pdf Download.
The SC is enabled by pressing the front panel scope key. Do not remove from this master document. The factory default is 4.